ISO-80369-7 Fig C.2 Male Reference Luer Slip Connector
ISO 80369-7 FIGURE C.2 (formerly ISO 594/1 Fig 5) MALE REFERENCE LUER SLIP CONNECTOR
ISO 80369-7:2016 Foreword
ISO 80369-7 cancels and replaces the first edition ISO 594-1:1986 and the second edition of ISO 594-2:1998, clauses, subclauses, tables, figures, and annexes of which have been consolidated and technically revised. Changes...
ISO 80369-7:2021 Foreword
This second edition cancels and replaces the first edition (ISO 80369-7:2016), which has been technically revised. Changes...
Compatibility/Interchangeability Between ISO 594/1 Figure 5 and ISO 80369-7 Figure C.2
The ISO 594/1 Figure 5 Male Reference Conical Fitting is replaced by ISO 80369-7 Fig C.5 Male Reference Luer Slip Connector. The ISO 80369-7 Figure C.2 is not identical to the ISO 594/1 Figure 5 Reference Fitting, so it cannot be certified as ISO 80369-7 Figure C.2. The good news is that the ISO 80369-7 Figure C.2 Male Reference Luer Slip Connector is 100% backward compatible with the ISO 594/1 Figure 5 Reference Fitting; so it can be certified and used in ISO 594/1 designated testing.
For those who just need to know the full details...
The minimum conical length changed from the ISO 594/1 Figure 5 requirement of 7.5mm minimum; to the ISO 80369-7 Figure C.2 requirement of 10.5mm minimum. I think it is obvious to see that an ISO 80369-7 Figure C.2 Male Reference Luer Slip Connector with a certified minimum length of 10.5 will always meet the ISO 594/1 Figure 5 Male Reference Conical Fitting requirement of 7.5mm minimum.
Escape Clause: ISO 80369-7; Page 27; Annex C; Figure C.2; Note states: The minimum length of the female taper of 10,5 mm is required for testing non-interconnectable characteristics. A minimum length of the female taper of 7,5 mm may be used for the performance tests of Clause 6.
What they are saying here is that an ISO 594/1 Fig 5 Reference Fitting with a certified length of 7.5mm minimum can be used for all of the tests of ISO 80369-7 Clause 6 testing, but if you sent that Reference fitting out to be certified to match ISO 80369-7 Fig C.2 it will fail because it does not meet the 10.5mm minimum. This escape Clause will allow you to use your old ISO 594/1 Fig 5 Reference Fittings until they wear out or until you cannot convince the oversight party that it is allowed.
ISO 80369-7 Fig C.2 Male Reference Luer Slip Connector
The ISO 80369-7 Fig C.2 Male Reference Luer Slip Connector is used for testing Female Luer Connectors for: Leakage; Separation from axial load; Stress cracking; and Non-interconnectable characteristics in the ISO 80369-7; Clause 6 specified tests. Keeping with the requirements of ISO 80369-7; the Reference Connectors we sell are from hardened stainless steel. The ISO 80369-7 Fig C.2 Male Reference Luer Slip Connector has design details which are not specified in ISO 80369-7 and which allow connection to test equipment. These design details vary by manufacturer so be sure you understand the configuration before you order the Reference Connector. The ISO 80369-7 Fig C2 Male Reference Luer Slip Connector is an integral part of Luer needle and syringe testing.
This website is funded by the sale of screw thread and medical small bore connector measurement tools. It will help us continue to offer this data if you allow us to have the opportunity to sell our tools to your company. If you find this information useful, please work to get Gage Crib Worldwide Inc. on the bid list for future purchases of medical small bore connector measurement tools.
This data is provided for general information only. The intention
is to provide accurate information; regardless; errors may exist
in the supplied information. If accuracy is critical, base your
final decisions on the data provided in the current versions of the root documents: ISO594/1; ISO 594/2;ISO80369-7; and ISO80369-20 which are copyrighted documents. To purchase copies visit
an Authorized Reseller.
Original Posting: 3/20/2009
Last Revision: 9/17/2021
Error corrections in, or comments about, the above data can be sent to: email@example.com